12.1" color TFT touch screen
Windows style, Multi-channel display
Big font interface
ST and arrhythmia analysis
Drug dose calculation and titration table
OxyCRG dynamic view display
Respiration monitoring with apnea alarms
Alarm review, arrhythmia review
Patient info. Input
NIBP over pressure protection
Plug-in rechargeable battery, AC/DC available.
Supporting wire or wireless network.
Adopting Aiah(TM) ECG technology, anti-interference, be able to collect the weak ECG signal away from interference.
Adopting Surepeb(TM) NIBP technology, accurate measurement to high blood pressure patient and low blood pressure patient.
Nellcor(TM) OxiMax(TM) is the world leading SpO2 technology
Nellcor(TM) OxiMax(TM) delivers excellent performance,
even in the difficult monitoring conditions of low perfusion and signal interference, including patient motion.
Collaborate with Respironics, Plug and Play EtCO2 monitoring.
CAPNOSTAT mainstream sensor for optimal performance in monitoring intubated patient. Small, durable and lightweight mainstream sensor provides accurate
and reliable monitoring for all intubated patients from neonates to adults. Loflo sidestream sensor for monitoting non-intubated patient.
Flexible, compact CO2 sensor provides consistent and reliable monitoring of adult, pediatric and neonatal patients.
Multi Mounting Solutions
Lead Mode 5 Leads (R, L, F, N, C or RA, LA, LL, RL,V)
Lead selection I, II, III, avR, avL, avF, V
Wave form 2 ch
Lead mode 3 Leads ( R, L, F or RA, LA, LL)
Lead selection I, II, III,
Wavef orm 1 ch
Gain 2.5mm/mV, 5.0mm/mV, 10mm/mV,20mm/mV, auto
HR and Alarm Range Adult 15 ~ 300 bpm
Neo/Ped 15 ~ 350 bpm
Accuracy ±1% or±1bpm, which is greater
Resolution 1 bpm
Sensitivity > 200 (uVp-p)
Differential Input Impedance > 5 MΩ
CMRR Monitor > 105 dB
Operation > 105 dB
Diagnosis > 85 dB
Electrode offset potential ±300mV
Leakage Current < 10 uA
Baseline Recovery < 3 S After Defi.
ECG Signal Range ±8 m V ( Vp-p )
Bandwidth Surgery 1 ~ 20 Hz
Monitor 0.5 ~ 40 Hz
Diagnostic 0.05 ~ 130 Hz
Calibration Signal 1 (mVp-p), Accuracy:5%
ST Segment Monitoring Range Measure and Alarm -2.0 ~ +2.0 mV
ARR Detecting Type: ASYSTOLE, VFIB/VTAC, COUPLET, BIGEMINY, TRIGEMINY, R ON T, VT>2,BRADY, MISSED BEATS,
PNP, PNC,Alarm Available,Review Available
|Method ||Impedance between R-F(RA-LL)|
|Differential Input Impedance ||I > 2.5 MΩ|
|Measuring Impedance Range ||I 0.3~3 Ω|
|Base line Impedance Range || 200~4 KΩ|
|Bandwidth ||I 0.1~2.5Hz|
|Resp. Rate Measuring and Alarm Range ||Adult 0 ~ 120 Brpm Neo/Ped 0 ~ 150 Brpm Resolution 1 Brpm Accuracy ±2 Brpm|
|Apnea Alarm ||10 ~ 40 S|
Mode Manual, Auto, STAT
Measuring Interval in AUTO Mode 1, 2, 3, 4, 5, 10, 15, 30, 60, 90,120, 180, 240,480 (Min)
Measuring Period in STAT Mode 5 Min
Pulse Rate Range 40 ~ 240 bpm
Alarm Type SYS, DIA, MEAN
Measuring and alarm range Adult Mode
SYS 40 ~ 270 mmHg
DIA 10 ~ 215 mmHg
MEAN 20 ~ 235 mmHg
SYS 40 ~ 200 mmHg
DIA 10 ~ 150 mmHg
MEAN 20 ~ 165 mmHg
SYS 40 ~ 135 mmHg
DIA 10 ~ 100 mmHg
MEAN 20 ~ 110 mmHg
Accuracy Maximum Mean error ±5mmHg
Maximum Standard deviation ±8mmHg
Overpressure Protection Adult Mode 297±3 mmHg
Pediatric Mode 240±3 mmHg
Neonatal Mode 147±3 mmHg
Measuring Range 0 ~ 100 %
Alarm Range 0 ~ 100 %
Resolution 1 %
Accuracy 70% ~ 100% 2 %
0% ~ 69% unspecified
Actualization intervalabout 1 Sec.
Alarm Delay 10 Sec.
Pulse Rate Measuring and Alarm Range 20~300 bpm
Measuring and Alarm Range 0 ~ 50 ºC
Resolution 0.1 ºC
Accuracy ±0.2 ºC
Actualization intervalabout 1 Sec.
Average Time Constant < 10 Sec.
Our Certificates:Why choose us:we got production and business certification from Chengdu food and Drug Administration
|Label ||ART, PA, CVP, RAP, LAP, ICP, P1, P2|
|Measuring and alarm range || ART 0 ~ 300 mmHg PA -6 ~ 120 mm Hg CVP/RAP/LAP/ICP-10 ~ 40 mm Hg P1 / P2-10 ~ 300 mmHg|
|Press Sensor Sensitivity || 5 uV/V/mmHg Impedance 300-3000Ω|
|Resolution || 1 mmHg|
|Accuracy || ±2% or 1mmHg which is greater|
|Actualization interval || about 1 secretary|
|Standard|| ECG, Respiration, NIBP, SpO2, Pulse Rate, Temperature|
|Optional|| Nellcor SpO2, EtCO 2, IBP, Thermal Recorder|
we got business license from Chengdu City Administration of industry and commerce
we got products certificate of registration and have checked in the website of SFDA
we have accepted the onsite check from SFDA every quarter
we finished the training of laws and regulations in medical equipments field
We have well-established system about material purchasing, employee training,store management,after-sales service process.etc
We adheres the beliefs of "Support Healthy Life", and our company spirits are quality, service,trust,thanksgiving;